Fluorophos ALP Diagnostic Test – Pipette Test 1
1/ Check the calibration of the 75µl sample pipette using the “calibration gauge” which is situated in the end of the pipette (see below).
2/ If the calibration is out re-calibrate the pipette. Adjustment is obtained by loosening the calibration set screw on the side of the pipette. Set the plunger depth using the calibration gauge then re-tighten the set-screw. Return the gauge to its storage position at the end of the pipette.
3/ Replace the pipette tip.